# Two-Point and Four-Point Methods for Measuring Small Resistances

 Difficulty Time Required Short (2-5 days) Prerequisites To do this project, you should be familiar with Ohm's Law, and with the basics of using a digital multimeter. Material Availability Readily available Cost Average (\$50 - \$100) Safety No issues

## Abstract

Measuring the value of a resistor with an ohmmeter is pretty simple. You connect the meter to the resistor, and read off the measurement from the meter. But what if the resistance you want to measure is very low? This project shows you how to use a four-point resistance measurement method to measure low resistance values.

## Objective

The goal of this project is to determine which method for measuring small resistances (<1 kohm) is best: the two-point or four-point resistance measurement method?

## Credits

Written by Charlie Zhai,

Edited by Andrew Olson, Ph.D., Science Buddies

Sources

### MLA Style

Science Buddies Staff. "Two-Point and Four-Point Methods for Measuring Small Resistances" Science Buddies. Science Buddies, 28 June 2014. Web. 3 Aug. 2015 <http://www.sciencebuddies.org/science-fair-projects/project_ideas/Elec_p025.shtml>

### APA Style

Science Buddies Staff. (2014, June 28). Two-Point and Four-Point Methods for Measuring Small Resistances. Retrieved August 3, 2015 from http://www.sciencebuddies.org/science-fair-projects/project_ideas/Elec_p025.shtml

## Share your story with Science Buddies!

Last edit date: 2014-06-28

## Introduction

You probably have learned how to measure the resistance of a copper wire by using a multimeter. Ohmmeter measurements are normally made with just a two-point measurement method (one probe on each of the two resistor leads). However, when measuring very small resistances, in the milli- or micro-ohm range, the two-point method is not satisfactory because test contact resistance becomes a significant factor. A similar problem occurs when making ground mat resistance tests, because long lead lengths (up to 1000 feet) are used. Here also, the lead resistance, due to long lead length, will affect the measurement results. The four-point resistance measurement method eliminates lead resistance or contact resistance. In this project, we will show you a scientific way to accurately measure resistance by minimizing the contributon of contact resistance.

The following description, from Tony Kuphaldt's website, AllAboutCircuits.com (Kuphaldt, 2003), explains:

• what contact resistance is,
• why it poses a problem when measuring small resistances, and
• how the four-point method for measuring resistance provides better accuracy for measuring small resistances.

"Suppose we wished to measure the resistance of some component located a significant distance away from our ohmmeter. Such a scenario would be problematic, because an ohmmeter measures all resistance in the circuit loop, which includes the resistance of the wires (Rwire) connecting the ohmmeter to the component being measured (Rsubject):

"Usually, wire resistance is very small (only a few ohms per hundreds of feet, depending primarily on the gauge (size) of the wire), but if the connecting wires are very long, and/or the component to be measured has a very low resistance anyway, the measurement error introduced by wire resistance will be substantial.

"An ingenious method of measuring the subject resistance in a situation like this involves the use of both an ammeter and a voltmeter. We know from Ohm's Law that resistance is equal to voltage divided by current (R = V/I). Thus, we should be able to determine the resistance of the subject component if we measure the current going through it and the voltage dropped across it:

"Current is the same at all points in the circuit, because it is a series loop. Because we're only measuring voltage dropped across the subject resistance (and not the wires' resistances), though, the calculated resistance is indicative of the subject component's resistance (Rsubject) alone.

"Our goal, though, was to measure this subject resistance from a distance, so our voltmeter must be located somewhere near the ammeter, connected across the subject resistance by another pair of wires containing resistance:

"At first it appears that we have lost any advantage of measuring resistance this way, because the voltmeter now has to measure voltage through a long pair of (resistive) wires, introducing stray resistance back into the measuring circuit again. However, upon closer inspection it is seen that nothing is lost at all, because the voltmeter's wires carry miniscule current. Thus, those long lengths of wire connecting the voltmeter across the subject resistance will drop insignificant amounts of voltage, resulting in a voltmeter indication that is very nearly the same as if it were connected directly across the subject resistance:

"Any voltage dropped across the main current-carrying wires will not be measured by the voltmeter, and so do[es] not factor into the resistance calculation at all. Measurement accuracy may be improved even further if the voltmeter's current is kept to a minimum, either by using a high-quality (low full-scale current) movement and/or a potentiometric (null-balance) system.

"This method of measurement which avoids errors caused by wire resistance is called the Kelvin, or 4-wire method, or 4-point method. Special connecting clips called Kelvin clips are made to facilitate this kind of connection across a subject resistance:

"In regular, 'alligator' style clips, both halves of the jaw are electrically common to each other, usually joined at the hinge point. In Kelvin clips, the jaw halves are insulated from each other at the hinge point, only contacting at the tips where they clasp the wire or terminal of the subject being measured. Thus, current through the 'C' ('current') jaw halves does not go through the 'P' ('potential,' or voltage) jaw halves, and will not create any error-inducing voltage drop along their length: (Kuphaldt, 2003)"

## Terms and Concepts

To do this project, you should do research that enables you to understand the following terms and concepts:

• resistance,
• contact resistance,
• 2-point resistance measurement,
• 4-point Kelvin resistance measurement.

## News Feed on This Topic

, ,
Note: A computerized matching algorithm suggests the above articles. It's not as smart as you are, and it may occasionally give humorous, ridiculous, or even annoying results! Learn more about the News Feed

## Materials and Equipment

To do this experiment you will need the following materials and equipment:

• 4 12-inch lengths of insulated electric wire (copper) for 4-wire resistance measurement,
• 4 banana plugs (for connecting to probe wires to multimeters),
• 4 alligator connectors (for connecting probe wires to resistors),
• 2 AA batteries (1.5v each),
• holder for 2 AA batteries,
• two digital multimeters, available at Amazon.com,
• a selection of standard 1/4 watt resistors for higher-resistance comparison (e.g., 3 ohm, 30 ohm, 300 ohm, 3000 ohm),
• any conductor you can find that you think has small resistance. For example, try a dime, a segment of copper pipe (used in plumbing), etc. The smaller resistance they have, the more advantage the 4-point test has over the 2-point test.

## Experimental Procedure

Note Before Beginning: This science fair project requires you to hook up one or more devices in an electrical circuit. Basic help can be found in the Electronics Primer. However, if you do not have experience in putting together electrical circuits you may find it helpful to have someone who can answer questions and help you troubleshoot if your project is not working. A science teacher or parent may be a good resource. If you need to find another mentor, try asking a local electrician, electrical engineer, or person whose hobbies involve building things like model airplanes, trains, or cars. You may also need to work your way up to this project by starting with an electronics project that has a lower level of difficulty.

### Making 4-Point Resistance Measurements

1. Strip the ends of the 1-foot wires to expose a short segment of bare wire.
2. Solder an alligator connector to one end of each wire.
3. Solder a banana plug connector to the other end of each wire.
4. Label two of the wires as A1 and A2 (label both the plug end and the alligator clip end).
5. Label the other two wires as B1 and B2 (again, labeling both the plug end and the alligator clip end).
6. Connect the two meters, wires and battery as shown in Figure 1, below. Make sure that the ammeter is initially set to its highest DC current range (you can later decrease the current range, if necessary).

Figure 1. Diagram for making 4-wire resistance measurements using two digital multimeters.
7. Read the ammeter and voltmeter data.
9. Repeat the measurement for each of the resistors and conductors you are testing.

### Making 2-Point Resistance Measurements

1. Disconnect leads A1 and B1 (refer to Figure 1, above).
2. Change the setting of multimeter at the place of voltmeter to Ohms (resistance) measurement.
3. Directly measure the resistance, Rsubject.
4. Repeat the measurement for each of the resistors and conductors you are testing.

### Questions

1. Compare the values you measured for Rsubject with each method. Over what range of resistance values do the methods produce similar results?
2. Over what range of resistance values do the methods produce different results?
3. Can you explain the differences?

## Variations

• What do you think would happen if you used 2-foot long pieces of wire to make your 4-wire connectors?
• Investigate the temperature dependence of resistance by repeating the measurements at different temperatures. You can use a hair dryer to heat up the resistors and conductors (or put them in the refrigerator or freezer for awhile). Keep in mind that the metals are normally very good thermally conductive material, so their temperatures can change rapidly.
• For a slightly more advanced project, see the article by Bob Nuckolls in the Bibliography (Nuckolls, 2004) on building a simple circuit for making accurate 4-wire resistance measurements with a single multimeter.

## Share your story with Science Buddies!

The Ask an Expert Forum is intended to be a place where students can go to find answers to science questions that they have been unable to find using other resources. If you have specific questions about your science fair project or science fair, our team of volunteer scientists can help. Our Experts won't do the work for you, but they will make suggestions, offer guidance, and help you troubleshoot.

## If you like this project, you might enjoy exploring these related careers:

### Electrician

Electricians are the people who bring electricity to our homes, schools, businesses, public spaces, and streets—lighting up our world, keeping the indoor temperature comfortable, and powering TVs, computers, and all sorts of machines that make life better. Electricians install and maintain the wiring and equipment that carries electricity, and they also fix electrical machines. Read more

### Precision Instrument & Equipment Repairer

One of the basic truths in the universe is that objects tend to go from a state of higher organization to a state of lower organization over time. In other words, things break down, and when those things are precision instruments or equipment, they require the services of very specialized technicians to restore them to their working order. Precision instrument or equipment technicians often combine a love of music, medicine, electronics, or antiques with delicate mechanical repair work. Read more

### Electrical Engineering Technician

Electrical engineering technicians help design, test, and manufacture electrical and electronic equipment. These people are part of the team of engineers and research scientists that keep our high-tech world going and moving forward. Read more

## News Feed on This Topic

, ,
Note: A computerized matching algorithm suggests the above articles. It's not as smart as you are, and it may occasionally give humorous, ridiculous, or even annoying results! Learn more about the News Feed

## Looking for more science fun?

Try one of our science activities for quick, anytime science explorations. The perfect thing to liven up a rainy day, school vacation, or moment of boredom.