Measuring Sugar Content of a Liquid with a Laser Pointer
Difficulty  
Time Required  Short (25 days) 
Prerequisites  None 
Material Availability  Readily available 
Cost  Average ($40  $80) 
Safety  Adult supervision recommended. Even lowpower lasers can cause permanent eye damage. Please carefully review and follow the Laser Safety Guide. 
Abstract
Did you know that you can figure out how much sugar is in a liquid without ever tasting it? In this science fair project, you will learn how to measure the concentration of sugar dissolved in a liquid by using a laser pointer, a hollow prism, and some physics. You will discover how refraction, or the bending of light, is the key to measuring the sugar content of a liquid with a laser pointer.Objective
Determine the sugar concentration of a liquid you can see through, like water or fruit juice, by measuring the solution's index of refraction.
Credits
Andrew Olson, Ph.D, Science Buddies.
Edited by Rose Leopold and Terik Daly, Science Buddies.
Information on making the hollow prism for this project came from:
 Edmiston, M.D., 2001. "A Liquid Prism for Refractive Index Studies," Journal of Chemical Education 78(11):14791480.
Cite This Page
MLA Style
APA Style
Share your story with Science Buddies!
I Did This Project! Please log in and let us know how things went.Last edit date: 20141018
Introduction
You may have seen an odd "bending" effect when you put a straw (or pencil) in a glass of water: the straw appears to bend at the boundary between the air and the water. Compare the two images in Figure 1 and see if you notice anything different between them.
The straws in Figure 1, above, seem to bend because of a phenomenon called refraction, the bending of a wave as it passes from one material into another. Waves travel at different speeds through different mediums (such as air and water), and this speed difference makes the wave refract when it passes from one material into another. As light waves travel from the water into the air, the wave refracts, which makes the straws in Figure 1, above, look bent or broken.
Can you see the refraction is different in the two parts of Figure 1, above? This is because one glass is filled with a denser solution than the other. The denser the medium, the slower the wave travels through the medium. The speed at which a light wave travels through a medium is quantified in the index of refraction, n, of that medium. It represents the ratio between the speed of light in a vacuum and the speed of light in the medium of interest. For example, the index of refraction of air is 1.00028, so light travels 1.00028 times faster in vacuum than it does in air.
The amount of refraction, or how much a light wave bends when it travels from one medium to another, is related to the indices of refraction by a mathematical formula called Snell's Law. We will use Figure 2, to understand Snell's Law. Figure 2 shows a ray of light (the red line in Figure 2) passing from air (white) to water (blue). Snell's Law uses the angle of a light ray entering a material, called the angle of incidence, θ_{1}, and the indices of refraction of the two materials (air and water, in this case) to calculate how much the light ray will refract as it passes from one material into the other. "How much the light refracts" is expressed mathematically as an angle called the angle of refraction, θ_{2}. Note that both the angle of incidence and the angle of refraction are measured from a line perpendicular to the surface the light interacts with. This line is called the surface normal, or simply, the normal (dashed gray line in Figure 2).
In equation form, Snell's Law looks like this:
Equation 1:
=
Index of refraction of material 2 × sin(angle of refraction)
n_{1} sin θ_{1}= n_{2} sin θ_{2}
 n_{1} = index of refraction of material 1 (no units, since it is a ratio)
 θ_{1} = angle of incidence (degrees or radians)
 n_{2} = index of refraction of material 2 (no units, since it is a ratio)
 θ_{2} = angle of refraction (degrees or radians)
Snell's Law tells us that the greater the difference between the indices of refraction of two materials, the more the light bends. If dissolving sugar in water creates a solution with a higher density than water alone, can you tell which glass in Figure 1, above, contains plain water and which one has sugar water?
If a light wave would travel through a prism filled with water, can you draw how the wave would be refracted?
In this science project, you will use the physics of refraction to measure the sugar content of a clear liquid solution (e.g. apple juice or a clear soda drink) using a laser pointer and a hollow prism. Figure 3, shows a picture of the basic setup of the experiment.
When there is no liquid in the prism, the laser will seem to shine straight through to a wall. See Figure 4. Refraction does happen as the laser beam crosses the prism walls, but the prism walls are so thin that we cannot see the refraction.
But, when the prism is filled with liquid, the laser is refracted twice, as shown in Figure 5.
Using the setup shown in Figures 3, 4, and 5, you will make measurements that allow you to calculate the angle of minimum deviation, θ_{md}. The angle of minimum deviation is the smallest possible angle between the incident ray (red line) and emergent ray (orange line), and it occurs when the light passing through the prism (solid blue line) is parallel to the base of the prism, as shown in Figure 5, above.
From θ_{md}, you can calculate the index of refraction using Equation 2, the index of refraction equation. You can derive Equation 2 by applying Snell's Law twice (once for each time the light wave changes mediums), and then using algebra, trigonometric identities, and the fact that the prism is equilateral.
Equation 2:
 n = index of refraction of solution (no units, since it is a ratio)
 n_{air} = index of refraction of air (no units, since it is a ratio)
 θ_{md} = angle of minimum deviation (degrees or radians)
 θ_{p} = apex angle of prism (degrees or radians)
Equation 2, above, looks complicated at first, but it is actually not so bad. θ_{md} is the angle of minimum deviation, which you will measure (we will show you how in the Experimental Procedure). θ_{p} is the apex angle of the prism, or the angle between the sides of the prism. Since the prism is an equilateral triangle, the apex angle is 60°. In Equation 3, we have substituted 60° for θ_{p}.
Equation 3:
In Equation 4, we have added the numerical value of the index of refraction of air (n_{air} = 1.00028). The sine of 30° is 0.5, and that is added in Equation 4.
Equation 4:
Finally, we simplify the numerical terms to produce Equation 5, which is the one you will use.
Equation 5:
 n = index of refraction of solution (unitless, since it is a ratio)
 θ_{md} = angle of minimum deviation (degrees)
To use equation 5, above, plug in your measured value for θ_{md}, add 60°, and multiply the result by 0.5. Then take the sine of the result, and multiply by 2.00056, and you will have the desired index of refraction.
Terms and Concepts
To do this project, you should do research that enables you to understand the following terms and concepts:
 Refraction
 Index of refraction
 Snell's Law
 Angle of incidence
 Angle of refraction
 Density
 Prism
 Angle of minimum deviation
Questions
 Do different wavelengths of light refract the same amount?
 Why would you use a laser in this science project, instead of another source of light? Remember that a laser has exactly one wavelength of light.
Bibliography
A simple summary of Snell's Law (the "plug in the numbers and calculate" version required for this project):
 Kaiser, P. (2005). "Snell's Law," The Joy of Visual Perception. Retrieved July 13, 2012, from http://www.yorku.ca/eye/snell.htm.
A fairly comprehensive tutorial that builds an intuitive understanding of Snell's Law using high schoollevel math:
 Henderson, T. (2004). "The Mathematics of Refraction, Snell's Law," The Physics Classroom. Retrieved July 13, 2012, from http://www.physicsclassroom.com/Class/refrn/U14L2b.cfm.
This reference discusses prisms and the angle of minimum deviation:
 Nave, R. 2010. "Prisms," Hyperphysics. Retrieved August 293, 2012, from http://hyperphysics.phyastr.gsu.edu/hbase/geoopt/prism.html.
Materials and Equipment
 Hollow prism; like this one from Amazon.com. or, you can make your own, following the directions in the Procedure
 To make your own hollow prism, you will need:
 1" x 3" glass microscope slides (4)
 Diamond scribe or glass cutter
 Ruler (with metric units)
 Electrical tape
 Epoxy glue (either 5minute or 30minute epoxy)
 Toothpicks
 Laser pointer
 Cardboard (a few small pieces; the flaps from a cardboard box work well)
 Masking tape
 Tape measure
 Paper, 8.5 inches by 11 inches
 Pencil
 Piece of string (~ 4 feet)
 Sugar
 Water
 Graduated cylinder (100 mL) or liquid measuring cup with metric units (1)
 Stirrer (tall enough to reach bottom of graduated cylinder or liquid measuring cup)
 Gram scale, such as the Fast Weigh MS500BLK Digital Pocket Scale, 500 by 0.1 G, available from Amazon.com
 Calculator with trigonometric functions (sine, arctangent), like ecalc.com.
Share your story with Science Buddies!
I Did This Project! Please log in and let us know how things went.Experimental Procedure
Laser Pointer Safety
Making the Prism from Microscope Slides
Note: If you are using a premade hollow prism, skip ahead to "Measuring the Index of Refraction of a Liquid".
 The goal is to make an equilateral prism that can hold liquid. It will be constructed from microscope slides and epoxy. See Figure 6, for an overview of how to do this. Each step is detailed in the procedure that follows.
 Put a piece of black electrical tape across the face of the microscope slide, as shown above (Figure 6a). The tape should hang over the edge by about 12 centimeters (cm).
 Score the other side of the microscope slide with a diamond scribe or glass cutter into three equal pieces, as shown above (Figure 6a). Use a straightedge to guide the diamond scribe. The two scribe lines should be one inch apart and perpendicular to the long edge of the slide. (If desired, before scribing you can mark the positions for the scribe lines with marker. The marker can later be cleaned off with a small amount of rubbing alcohol on a paper towel.)
 Now you will break the glass along the scribe lines. Hold the slide on either side of the first scribe line and bend the glass toward the taped side. Bend just enough to break the glass. Repeat for the second scribe line (Figure 6b).
 Now bend the glass away from the tape, allowing the tape to stretch (Figure 6c). Continue bending until the triangle closes.
 Place the prism on a flat surface to align the bottom edges. Use the overhanging tape to secure the prism in this configuration (Figure 6d).
 Adjust the edges of each face so they align correctly. At each apex of the prism, the inside edges should be in contact along their entire vertical length.
 Follow the manufacturer's instructions for mixing the epoxy cement (usually you mix equal amounts from each of two tubes). Use a toothpick to apply epoxy to the inside corners and along the inside seams between the faces of the prism to glue the three faces together (Figure 6e). The seams need to be watertight, but keep the epoxy in the seams and away from the faces of the prism. Keep the bottom surface flat and allow the epoxy to set. Remove the tape once the epoxy is dry.
 When the epoxy in the seams has set firmly, mix up fresh epoxy and use a toothpick to apply it to the bottom edge of the prism. Glue the prism to a second microscope slide as shown (Figure 6f). The bottom edge needs to be watertight, but keep the epoxy away from the faces of the prism.
 Allow the epoxy to set overnight. Your prism is ready for use.
Measuring the Index of Refraction of a Liquid
Note: Do this project in an area where you can put a table close to a flat wall or window, and where taping paper to the wall or window is allowed.
 Figure 7, is a diagram of the setup you will use for measuring the index of refraction of a liquid. This is what the setup would look like if you were looking down on it from above. (Note that the diagram is not to scale.)
 Lay the laser pointer on a table. The laser pointer should be set up so its beam (dotted red line in Figure 7) is perpendicular to a nearby wall.
 Lay a piece of paper in front of the laser pointer. Tape it securely to the table. The paper will be used to mark where the laser beam enters and exits the prism.
 Place the prism on top of the paper, a few centimeters in front of the laser pointer. One of the prism's triangular faces should be resting on the paper, as shown in Figure 8. Using a pencil, trace around the prism's base. If you move the prism, always return it to this location before rotating it, if needed, as explained in step 12.
 Adjust the height of the laser pointer with pieces of cardboard until the laser's beam hits about halfway up the side of the prism. See Figures 8 and 9.

Tape the cardboard to the table, and then tape the laser pointer to the cardboard. Make sure that neither the cardboard nor the laser pointer can move. If the laser pointer's position changes, your measurements will not be accurate.
 Troubleshooting Tip: If you have two or more sheets of cardboard stacked together, you may need to tape the pieces of cardboard together so that they do not slip.
 Tape a big piece of paper to the wall in front of the laser pointer. You will use this paper to mark where the laser beam hits the wall.
 To measure the angle of minimum deviation, θ_{md}, which you will use to calculate the index of refraction of the liquids that you test, you need to mark several points and measure the distances between some of these points. Figure 10, is a more detailed view of the prism and wall. It shows all the points you need to mark in order to measure the angle of minimum deviation, θ_{md}. The procedure below explains how to mark these points and determine the angle of minimum deviation, θ_{md}.

When the prism is empty (filled only with air), placing it in the laser's path should not divert the
beam. Turn the laser on, and mark the spot where the beam hits the paper taped to the wall. Mark this as
point b (point b in Figure 10).
 Troubleshooting Tip: Before testing a new solution, turn on the laser and shine it through an empty prism to make sure that the laser beam still hits point b. If the laser beam no longer hits point b, your measurements will not be accurate. Adjust the laser's position, if necessary, until the undiverted beam hits point b.
 With the prism empty, mark where the beam enters the prism on the paper the prism is sitting on (point d in Figure 10). Label it point d.
 With the prism still empty, mark where the laser beam exits the prism on the paper the prism is sitting (point e in Figure 10). Label it point e.
 Turn off the laser. Fill the prism with plain water. If you moved the prism to fill it with water, return it to the outline you made on the piece of paper. Turn the laser back on.

Rotate the prism so that the path of the refracted beam within the prism (solid blue line
from d to f in Figure 10) is parallel with the base of the prism, the side of the prism that has no
laser beam hitting it.
 Troubleshooting Tip: A pinch of nondairy creamer in the liquid can help you see the beam within the prism, and should not have a significant effect on the index of refraction of the liquid. Or, if you do not have nondairy creamer, take a straight edge and line it up with the laser beam's entrance and exit points (as seen when looking at the prism from a top view). Rotate the prism until the straightedge connecting those two points is parallel to the side of the prism that the laser beam does not hit.
 When the prism is rotated correctly (as described in step 13), mark the position where the emerging beam hits the paper taped to the wall (point a in Figure 10). Label it point a.
 On the paper on the table, mark the point where the beam emerges from the prism (point f in Figure 10). Label it point f.
 Now you can move the prism aside. Leave the papers taped in place.
 Use a ruler to draw a line from point d to point e. This marks the path of the undiverted beam.
 Next, extend a line from point a (on the wall) through point f (on the table). To do this, stretch a string from point a so that it passes over point f. Mark the point where the string crosses the line between d and e. This is point c.
 Measure the distance between points a and b, and record it in your lab notebook. This is distance x (see Figure 10).
 Measure the distance between points b and c, and record it in your lab notebook. This is distance L (see Figure 10).
 The distances you have measured define the angle of minimum deviation, θ_{md}. The ratio x/L is the tangent of the angle of minimum deviation, θ_{md}. To calculate the angle, use your calculator (or ecalc.com) to find the arctangent of x/L. (The arctangent of x/L means "the angle whose tangent is equal to x/L.") Record the angle and its units (radians or degrees) in your lab notebook.
 Now that you have the angle of minimum deviation, you can use Equation 5, to calculate the index of refraction, n, of the liquid in the prism.
Equation 5:
 n = index of refraction of solution (unitless, since it is a ratio)
 θ_{md} = angle of minimum deviation (degrees)
 Check that your setup is working correctly by measuring the index of refraction of plain water using steps 9 through 22 of this procedure. You should get an index of refraction of about 1.334.
Standard Sugar Solutions for Comparison

You will make three sugar water solutions, using the amounts of sugar and water shown in Table 1. Use the gram scale to weigh out the appropriate amount of sugar.
 Troubleshooting Tip: Using warm water will help the sugar dissolve more quickly.
Desired concentration (% mass/volume) 
Amount Sugar (g)  Amount Water (mL) 
10  10  90 
20  20  80 
30  30  70 
 Mix each of the solutions in Table 1 in a graduated cylinder or liquid measuring cup with metric units, using a stirring rod to dissolve the sugar. Once the solutions are made and the sugar is completely dissolved, set aside the 20% and 30 % solution and fill the prism with as much of the 10% sugar solution as possible.
 Measure the index of refraction of the 10% sugar solution (following the steps in the "Measuring the Index of Refraction of a Liquid" section of the procedure, above). Repeat your measurements 4 more times for the 10% sugar solution, for a total of 5 replicates. Average your results.
 Empty the prism and rinse it with plain water. Then repeat step 3 using 20% and then 30% sugar solutions.
 Now measure the index of refraction of a solution with unknown sugar concentration (e.g., a clear soft drink or fruit juice). If you measure a carbonated beverage, make sure that there are no bubbles in the path of the laser (gently dislodge them from the side of the glass, if necessary).
 With the index of refraction of the unknown solution, combined with the data you have from your known sugar solutions, you should be able to estimate the sugar concentration of the unknown solution.
Troubleshooting
For troubleshooting tips, please read our FAQ: Measuring Sugar Content of a Liquid with a Laser Pointer.
Share your story with Science Buddies!
I Did This Project! Please log in and let us know how things went.Variations
 Compare the index of refraction of regular and diet soda. Is there a difference?
 Can you use index of refraction to measure the concentration of salt dissolved in water? Make salt solutions with different known concentrations and find out. If you live near a body of salt water, can you use this method to estimate the salt concentration of salt water samples from different locations? This would be especially interesting to measure where fresh and salt water meet, e.g., in a tidal estuary where a river or stream meets a bay or the ocean.
 Advanced. Slowly pour water containing a pinch of nondairy creamer over a layer of sugar crystals in the bottom of an aquarium, trying not to allow too much turbulence to develop in the water. Wait for an hour or two to allow a concentration gradient to form as the sugar crystals dissolve. Predict what will happen when a beam of light shines through the solution. Shine a laser pointer through the solution. Can you account for the path that the beam follows in the liquid?
Share your story with Science Buddies!
I Did This Project! Please log in and let us know how things went.Frequently Asked Questions (FAQ)
Your measured index of refraction may be off due to inaccuracies in data collection or problems in your calculations. If you have a value around 1.334, in the 1.32 to 1.34 range, your error is most likely from your measurements, but it is good to double check your calculations.
If your index of refraction is very far from 1.334 you probably have a problem in your calculations. You should redo your calculations (paying attention to order of operations and checking for consistent units). Also check that the experiment is set up correctly and that you are measuring the correct distances.
Ask an Expert
Related Links
If you like this project, you might enjoy exploring these related careers:
Physicist
Physicists have a big goal in mind—to understand the nature of the entire universe and everything in it! To reach that goal, they observe and measure natural events seen on Earth and in the universe, and then develop theories, using mathematics, to explain why those phenomena occur. Physicists take on the challenge of explaining events that happen on the grandest scale imaginable to those that happen at the level of the smallest atomic particles. Their theories are then applied to humanscale projects to bring people new technologies, like computers, lasers, and fusion energy. Read morePhysics Teacher
Our universe is full of matter and energy, and how that matter and energy moves and interacts in space and time is the subject of physics. Physics teachers spend their days showing and explaining the marvels of physics, which underlies all the other science subjects, including biology, chemistry, Earth and space science. Their work serves to develop the next generation of scientists and engineers, including all healthcare professionals. They also help all students better understand their physical world and how it works in their everyday lives, as well as how to become better citizens by understanding the process of scientific research. Read morePhotonics Technician
Do you enjoy watching cable television, texting on your phone, and surfing the Internet? Do you know anyone who has had eye surgery and been back to normal the next day? Many of the advances in telecommunications and medicine are due to laser and fiberoptic technology. This technology has led to devices that provide faster and richer communication, advanced surgeries, and faster healing times, as well as amazing robotics for manufacturing. But, as with all equipment, someone has to install and maintain it. That is what photonics technicians do. These professionals are responsible for building, installing, testing, and maintaining optical and fiberoptic equipment such as lasers, lenses, and optics systems. Photonics technicians contribute to the technology that has drastically changed how we communicate and how we live. Read moreLooking for more science fun?
Try one of our science activities for quick, anytime science explorations. The perfect thing to liven up a rainy day, school vacation, or moment of boredom.
Find an Activity